Led Dental Handpiece With Quick Coupling
|Standard air pressure||0.25-0.3Mpa|
|Ways of unloading burs||Latch Type|
Led Handpiece(Self Illimination) With Quick Coupling.
Chuck Type:Push button
Workign Air Pressure:0.22-0.30MPa(0.25MPa prefrred)
Atomization Pressure:Under 5.00Mpa(0.30MPa prefrred)
Supply Voltage:3.2-3.5VFree Load Speed:≥280,000R.P.M for torque
head ≥350,000R.P.M for standard head
Life Of Led Bulb:≥5000hours
Led Luminous Intensity:≥1500mcd
Speed Of Generator:18,000-22,000R.P.M
Led Color Temperature:3500K-4000K
Dental handpieces right use way:
1. The need to use a clean , dry , oil-free air compressors for
medical use ;
2. The air pressure between 2.3-2.8kgf / cm2;
3. Place the adapter with quick connector at the bottom of a
combination of dental handpieces , and confirm that the card lock
4. Press the pedal , handpiece operation and examine whether the
normal operation of the spray head . If everything is normal, you
can start using the handpiece.
After each patient use , maintenance should be done as follows
1. First remove the diamond burs , and clean up all the teeth on
dental handpieces precipitate ;
2. Use a metal needle to clean the spray holes and fumaroles ;
Note: The spray holes and diamond burs into 40 -degree angle ,
fumaroles and diamond burs to a 30 degree angle.
3. Place the clean lubricating oil is sprayed into the axial hole
1-2 seconds ( diamond burs is plugged holes ) ;
4. In the bottom of the handpiece were ( with / without quick
connector ) two kinds of styling , make the right amount of
lubricant is injected into the bottom ;
5. The handpiece with quick connector combination ( first determine
the water switch is turned off )
6. Replace the diamond burs , surrounded by white silk cloth
handpiece head ( make sure the needle does not touch the diamond
burs and white gauze , so as not to tie ) ;
7. pedal depressed for about 20 seconds , look at whether there is
in white gauze on any stain. If there are stains on the white gauze
, again, add oil , starting from item 4 continued routine
maintenance , know that white gauze until no stain ;
8. Finally, no more than 135 degrees Celsius temperature and high
Consists of handpiece head, handle, caudal thread, cartridge.
1.Air pressure 0.25~0.30mpa,must be filtrated.
2.Use high speed dental bur with shaft diameter
Φ1.59~1.60mm×21-23mm and roghtness less than 0.8um.
3.The amount of handpieces spray or water spray should be able to
4.Hard glass fiber,can be repeatedly sterilized in high temperautre
Only used for drilling, molar of stomatology clamping high-speed
HPA013QC Handpiece we Provide Three Kind Of Head For Choice
|Mini Head||Standard Head|
Connect the tubing(4 or 2 or 6 hole) if appears quick coupling
handpiece, insert the quick coupling head in to, installation as
Notice:1.Connect the required tubing according the model of each
Handpiece With Quick Coupling
2.Only connecting correctly,fixed tightly and closely, can be used.
1.Inset quick couping
Insert quick coupling into socket connection of the high speed
handpiece, push it forward,card tightly and firmly.
Notice:For quick coupling with decorative pattern, should be moving
back before push forward, the can be card tightly and firmly
2.Extract the quick coupling
Hold the handpiece body and quick coupling by hands and drag them
Notice:For quick coupling with decorative pattern, holding the
decorative patten quick coupling, drag it in to behing.
Note:Only connecting correctly, fixed tightly and closely, can be
1.Considering safety first, please be careful to use it with full
2.Be limited to dentists who held the dental treatment
3.Strictly forbid decomposition and reconstruction by unauthorized.
4.Please run point testing on outside of patients oral beforehand.
If feel anomalies, please stop using immediately and contact with
our after-sales center.
5.When appears abnormal during using, please stop it immediately
and contact with our after-sales center.
6.It will cause a high noise using a center-vibration or bended bur
and a handpiece of which bearing has been worn, A long time in this
state will be harm to people’s hearing. Forbid using a vibration or
bended bur, if appear high noise, please replace the shaft core
7.Do not use a bending , cracking, deformation and unqualified bur.
If using it, may cause a snap, or a flying off to injury accident
during operation process.
8.Please ensure the bur chuck be clean. If with contaminant inside
of it, may causes bur clamping weaken or shaft core vibration.
9.Please run the bur under the provision rotation-speed by bur
10.Do not crash, especially to avoid falling.
11.Please install a bur or a standard stick before using.
12.Aslo remember to install a bur or a standard stick when not in
13.Please unload the bur and handpiece when the running stopped in